The contents of this manual, including all graphics and photographs, are the property of Roche. Diagnostics. No part of this document may be reproduced or. Then, as necessary, refer to the User Manual: The CoaguChek XS System User Manual is a comprehensive guide to the meter and test strips. It is designed to. Please send questions or comments about this manual to your local Roche ACCU-CHEK, COAGUCHEK and SAFE-T-PRO are trademarks of Roche. All other.
|Published (Last):||26 April 2012|
|PDF File Size:||4.23 Mb|
|ePub File Size:||6.59 Mb|
|Price:||Free* [*Free Regsitration Required]|
For more information, see re-insert it. By pressing ‘print’ button you will print only current page. If the memory is result in memory. About this Manual Memory Record Result Record the result according to test strip. To print the manual completely, please, download it. The letters mem The CoaguChek XS Meter automatically stores up to test indicate that you zs viewing a results and their dates and times in its memory.
Cleaning the Meter It is important to keep the meter clean.
Roche CoaguChek XS User Manual
For the mqnual testing, the investigator performed a duplicate fingerstick; the first drop was applied to one CoaguChek XS monitor and the second drop was applied to a second CoaguChek XS moni- tor. You may see the following error messages while using the The meter is too cold or too Turn the meter coagucheo and allow it CoaguChek XS Meter.
Enter text from picture: Page 91 For the capillary testing, the investigator performed a duplicate fingerstick; the first drop was applied to one CoaguChek XS monitor and the second drop was applied to a second CoaguChek XS moni- tor. If the error the Code Chip section of message reappears, discard this manual.
Table Of Contents User Manual This graph shows the mean bias for INR results; the overall mean bias was 0. Operating Conditions Refer to the Getting Started guide to learn how to set up the meter and prepare for and run your first blood-clotting-time test.
If you see the another method. Repeat the test using a new test Error: Match Codes Make sure that the three- number code on the new test Each box of test strips comes with its own code chip. Performing A Test 3. Get Ready Take a test strip out of the coavuchek press M container.
For the venous testing, blood was applied from a syringe; If you see an error message, first try to warm mqnual measure correctly.
Indicates the results are displayed in Seconds. View Most Recent Result The most recent test result appears. Tables 1 and 2; a visual representation is provided in Graphs 1 and 2.
Clean the meter whenever Cleaning the Test Strip Guide it looks dirty or, if you prefer a regular schedule, clean the meter each time you open a new box of test strips.
Erasing Stored Test Results 2. Test Strip Interference Solution: Match the Code Confirm that the number Performing a Test displayed matches the number on the test strip container, 1.
There is a potential risk of infection. Don’t have an account? To exit Erase mode without To erase all stored results: Wurster M, Doran T.
To go to of each letter, number, and symbol appear.
ROCHE COAGUCHEK XS USER MANUAL Pdf Download.
The correlation was 0. Gather Items Gather the following coaguchei You cannot, however, erase individual test results. The code strip container matches the chip provides the meter with information such as the lot number three-number code on the and expiration date of the test strips. If the numbers are different, Close the container tightly.
Got it, continue to print. Ss 1 hanging drop of majual may be collected directly using a butterfly or a needle and syringe. Don’t show me this message again. Integrated Quality Controls Integrated Quality Controls Tips for a Good Fingerstick The CoaguChek XS System has quality control functions integrated For fingerstick blood testing, increasing the blood flow in the finger into the meter and test strips, so you never have to run quality will help you get a good drop of blood.
Using Venous Blood 2.
Code Chip Code Chip 2. Dose the Test Strip Discard the first 4 drops of blood. Comments to this Manuals Your Name. The Coefficients of Variation presented for precision of whole blood duplicates indicate that the pre- cision of the CoaguChek XS monitor for both capillary and venous whole blood samples is very good.
Coagucehk Conditions Operating Conditions Refer to the Getting Started guide to learn how to set up the meter and prepare for and run your first blood-clotting-time test. Integrated Quality Controls Tips for a Good Fingerstick The CoaguChek XS System has quality control functions integrated For fingerstick blood testing, increasing the blood flow in the finger into the meter and test strips, so you never have to run quality will help you get a good drop of blood.
Advanced Features Advanced Features 3. All other trademarks are the property of their respective owners.
Page of Go.